接收并检测质量流量控制器，并对其进行外观检查。 质量流量控制器信息录入系统，生成一个工艺单，匹配适当的标准和适用的校准数据。 如果需要，预校准数据将被记录并提供 (附加费用)。 质量流量控制器完全拆开、清洗 更换所有密封件并对质量流量控制器重新组装 beplay电流与硬件调整。 所有的电子部分/机械部分调整都需要满足客户的要求。 质量流量控制器测试是否漂移。 按原厂标准进行预检漏测试。 由校准系统自动出具COC证书 包装及运输 质量流量控制器在防静电包装袋中密封。 证书与登录再次进行装袋密封。 我们服务全部按高标准质量控制系统来进行。 我们的清洁和校准程序分为四个过程:(1)清洗，(2)组装，(3)校准，(4)装运 The mass flow controller is received and visually inspected for damage. The mass flow controller is logged into our computer, which generates a worksheet displaying the proper standard and applicable calibration data. If requested, pre-calibration data is taken and graphed per customer request (additional fee). The mass flow controller is disassembled and chemically and ultrasonically cleaned. All seals are replaced and the mass flow controller is re-assembled The sensor current and bypass adjustments are made. All electronic/mechanical adjustments required to meet the customer specifications are made. The mass flow controller is tested for long-term drift. Calibration is performed at each calibration point with a calibration cover installed. Pre-shipment leak testing to the manufacturer’s specifications is performed. A Certificate of Calibration summarizing the calibration data and repair report is generated for your analysis and records. Packaging and Shipping The mass flow controller is doubled bagged in anti-static sleeving, and sealed. (All removed parts are sealed and returned in a separate bag). The mass flow controller is then, double wrapped with bubble wrapping, boxed and packed with Styrofoam. We utilize a unique (100% in-process) Quality Control System. Our Clean & Calibration procedures are divided into (4) processes: (1) Cleaning, (2) Assembly, (3) Calibration and, (4) Shipping Each process is inspected for defects or faults before it is transferred to the next station.